Experimental Section & Acknowledgements
Acknowledgement of NSF Support - Instrument Purchase
"Purchase of the AVIII-600 NMR spectrometer used to obtain results included in this publication was supported by the National Science Foundation under Grant No. MRI 1040098."
"Any opinions, findings, and conclusions or recommendations expressed in this material are those of the author(s) and do not necessarily reflect the views of the National Science Foundation."
Acknowledgement of CIF Staff Assistance
"We wish to thank ISU Chemical Instrumentation Facility staff members <insert name(s) here> for training and assistance pertaining to the AVIII-600 results included in this publication."
Experimental Section & Publication Information
The AVIII-600 NMR spectrometer features a narrow bore 14.1 tesla superconducting magnet and an Avance III spectrometer console from Bruker Corporation (Billerica, MA and Karlsruhe, Germany). The spectrometer uses a BCU-I or a liquid nitrogen exchanger for low temperature experiments. The amount of additional experimental detail necessary will depend upon whether you are preparing a manuscript, a poster, a presentation, or including the information in your thesis.
The AVIII-600 is operated using Topspin 3.0. Every data file acquired with Topspin has associated files with many of the relevant parameters from the acquisition. This data file can be accessed by opening the associated files in a text editor, or through Topspin by loading the file and typing the command dpa<enter>. The dpa file typically does not contain information about the temperature or sample rotation speed at which the data was acquired.
For measurements on the AVIII-600, this data file may include:
- Pulse sequence
- Number of scans (ns)
- Acquisition time (aq)
- Pulse lengths in microseconds (typically conveted to kHz for publication - i.e. 5 us = 50 kHz pulse)
The spectrometer frequencies for many typical nuclei are as follows:
- 1H = 600.39 MHz
- 19F = 544.93 MHz
- 31P = 243.04 MHz
- 11B = 192.63 MHz
- 13C = 150.97 MHz
- 29Si = 119.28 MHz
- 2H = 92.16 MHz
- 15N = 60.84 MHz
There are two probes available for the AVIII-600:
- 2H/1H/BB BBFO SmartProbe capable of tuning to nuclei of 109Ag-19F on the broadband channel. (Autotuning probe.)
- 2H/13C/1H/BB inverse probe with a dedicated 13C channel and 109Ag - 19F range on the broadband channel. (Does not have autotune.)